Bubble Viscometer Procedure
Please be aware of the following accuracies when performing the test:
ASTM D 1545 Timer Method
The tube has three amber ring marks at 27, 100 and 108 mm from the bottom. Fill the tube up to the 100 mm line, insert the cork down to the 108 mm line and turn the tube bottom up. Turn the tube around, start the stop watch when the air bubble crosses the 27 mm line and stop when the bubble crosses the 100 mm line.
Guide to Viscosity Measurement
Viscosity is an important property of liquids to determine how they flow, spread, or level under certain conditions. It can be defined as a liquid’s internal resistance to flow. Being able to measure the viscosity will allow to predict the behavior in a process or an application. When applying a paint by roll application the paint viscosity will strongly influence the amount of paint applied per surface area, thereby determining both yield and hiding power. It is crucial for liquid products to undergo a viscosity measurement in incoming and outgoing QC.Read more